A clean stylus ball on a CMM or Equator is key to achieving repeatable metrology results. Contaminant build up on the
stylus ball from general use (especially when scanning) can lead to poor metrology results; in order to combat this,
frequent inspection and cleaning of the stylus ball will ensure that results are as accurate as possible.
This type of file requires a viewer, freely available from